|Year : 2013 | Volume
| Issue : 3 | Page : 348-351
Accuracy of Prader orchidometer in measuring testicular volume
TU Mbaeri, JC Orakwe, AM Nwofor, KC Oranusi, OO Mbonu
Department of Surgery, Nnamdi Azikiwe University Teaching Hospital, Nnewi Anambra State, Nigeria
|Date of Acceptance||21-Oct-2012|
|Date of Web Publication||14-Jun-2013|
T U Mbaeri
Department of Surgery, Nnamdi Azikiwe University Teaching Hospital, Nnewi Anambra State
Source of Support: None, Conflict of Interest: None
| Abstract|| |
Background: Seminiferous tubules comprise 80-90% of testicular mass. Thus, the testicular volume is believed to be an index of spermatogenesis. Therefore, accurate testicular volume is one way to assess testicular function.
Objective: To determine the accuracy of Prader orchidometer for measuring the testicular volume by comparing the resultant measurement with the actual testicular volume in humans.
Materials and Methods: The testicular volumes of 121 testes from 62 patients with prostate cancer (mean age 72.74 ± 9.38 years) were measured using Prader orchidometer before therapeutic bilateral orchidectomy. The actual testicular volumes were then determined by water displacement of the testis.
Results: The mean testicular volume of the 121 testes was 10.60 ± 3.5 ml and13.26 ± 5.2 ml for water displacement and Prader orchidometer measurements, respectively. A strong correlation was found between the actual testicular volume and volumes obtained by Prader orchidometer ( r = 0.926, P = 0.0001). The Prader orchidometer however, over-estimated the mean actual testicular volume by 2.66 ± 2.37 ml (25.10%).
Conclusion: The result of this study has shown that measuring the testicular volume by Prader orchidometer overestimates the actual testicular volume.
Keywords: Accuracy, prader orchidometer, testicular volume
|How to cite this article:|
Mbaeri T U, Orakwe J C, Nwofor A M, Oranusi K C, Mbonu O O. Accuracy of Prader orchidometer in measuring testicular volume. Niger J Clin Pract 2013;16:348-51
|How to cite this URL:|
Mbaeri T U, Orakwe J C, Nwofor A M, Oranusi K C, Mbonu O O. Accuracy of Prader orchidometer in measuring testicular volume. Niger J Clin Pract [serial online] 2013 [cited 2022 Nov 30];16:348-51. Available from: https://www.njcponline.com/text.asp?2013/16/3/348/113460
| Introduction|| |
The testes are responsible for the production of spermatozoa and testosterone in man. Approximately, 80-90% of testicular volume is made up of seminiferous tubules and germ cells. , Thus, a reduction in the number of these cells is manifested by reduction in testicular volume.  Reliable and accurate determination of testicular volume is of great benefit in evaluating patients with a variety of disorders affecting testicular growth, development, and function.
Studies in infertile men have also shown that testicular volume correlates with seminal fluid assay parameters and sex hormone values, just like the simple measurement of testicular length, width, and depth. ,, Total testicular volume by orchidometer (i.e., summation of right and left) of 30 ml and above is indicative of normal testicular function.  These findings underscore the great importance of testicular volume measurement in the management of male infertility. In line with this, one of the components of minimum full evaluation for male infertility is palpation of the testes and measurement of their sizes. 
In the management of adolescent varicocele, testicular volume measurement aids in deciding when to operate, as getting seminal fluid analysis could be seen to be psychologically and ethically incorrect. ,, Other important uses of testicular volume measurement are the monitoring of patients following varicocele ablation in children and adults, and orchidopexy for undescended testes. , It is also a vital tool in staging puberty, as it is the first clinical evidence of puberty,  and in making a diagnosis of hypogonadal hypogonadism and Klinefelter's syndrome. ,,,
Over the years, many instruments have been used in an attempt to accurately, reliably, and conveniently measure the testicular volume in vivo. These include rulers, tapes, vernier callipers, orchidometer, graphic models, and ultrasound scan. ,,,,,,,,,,,,,, Results with these instruments have been conflicting. So, we sought to critically assess the accuracy of Prader orchidometer which is cheap and can be made readily available in our clinics for routine testicular volume measurement.
| Materials and Methods|| |
This study was carried out over a period of 19 month, from June 2009 to December 2010. It is a hospital based cross-sectional prospective study of the testicular volume of patients with advanced prostate cancer who opt for bilateral orchidectomy, after counseling, as a form of hormone ablation therapy in our center.
Ethical approval was sought and obtained from the ethical committee of the hospital. All patients who had hydrocele, painful testis, and edematous scrotum were excluded. Also, those who did not give consent for the study were excluded. Patients who qualified for the study had their scrotum and contents examined, and the testicular volumes measured with a Prader orchidometer. The examinations were done in a warm room after application of a heating pad (we used hot water bottle) to the scrotum for about 5 min. The testes were then gently isolated and distinguished from the epididymis; and the scrotal skin stretched without compressing the testis. Using the orchidometer, a manual side-by-side comparison between the testis and the beads were made to identify the bead most similar in size to the testes; this indicated the testicular volumes.
Orchidectomies were then performed for the patients, tagging the right testis for identification and removing the epididymis by sharp dissection. The actual testicular volumes were measured by water displacement method using a measuring cylinder. All the results were recorded in the study proforma.
Statistical analysis was done using Statistical Package for Social Sciences (SPSS) version 16.0. Simple frequencies were determined for age and descriptive statistics for the testicular volume measurements. Test of significance for testicular volumes was done using paired sample t-test, and correlation was done using Pearson correlation coefficient.
| Results|| |
A total of 62 patients were studied. Fifty-nine patients had bilateral testes, two had only the right testis, and one had only the left testis. So, only 121 testes were used in this study.
The age ranged from 55 years to 92 years, with a mean age of 72.74 ± 9.38 years. The peak age group was in the age range 71-75 years, which had 18 patients (29%).
The mean testicular volume measured by Prader orchidometer was 13.26 ± 5.21 ml (range 6-25) whereas the mean actual testicular volume as measured by water displacement was 10.60 ± 3.51 ml (range 4.40-20.00) [Table 1].
|Table 1: Mean testicular volume measurement by Prader orchidometer and water displacement |
Click here to view
The Prader orchidometer overestimated the mean testicular volume by 2.66 ± 2.37 ml, that is by 25.10% [Table 2].
|Table 2: Paired samples test between orchidometer and water displacement |
Click here to view
This mean difference in testicular volume between Prader orchidometer and the actual testicular volume measured by water displacement was statistically significant by the paired t-test (P = 0.0001). Though the mean difference in testicular volume was statistically significant, the mean testicular volume measured by Prader orchidometer correlated strongly with the actual testicular volume with Pearson correlation coefficient r = 0.926 and P = 0.0001. The scatter diagram is as shown in [Figure 1].
|Figure 1: Scatterplot of prader orchidometer and water displacement (*R: Pearson correlation coefficient)|
Click here to view
| Discussion|| |
The mean age of patients in this study was 72.74 ± 9.38 years. This is similar to the mean age of 74.5 ± 7.5 years reported by Sakamoto et al.  who studied testicular volumes of 40 testes in 20 patients. A study by Ogunbiyi et al.  in Ibadan Nigeria on the incidence of prostate cancer in Nigeria gave the mean age of 71.4 years (variance 14.3). Thus, considering that this study was done in patients with prostate cancer as our study population this age range of 72.74 ± 9.38 years is not unexpected.
The mean actual testicular volume measured by water displacement was 10.6 ± 3.5 ml (4.4-20.0) in this study. This mean actual testicular volume is the same as what Hsieh et al.  got in a similar work in China. He got a volume of 10.6 ml. It is however different from that of Sakamoto et al.  who got a mean actual testicular volume of 9.3 ± 4.5 ml. This difference may be because of his study of a smaller number of testes (40) compared to the 121 testes we studied. However, Hsieh et al.  who got a similar mean volume studied only 30 testes, which is even smaller than what Sakamoto et al. studied. Therefore, the sample size alone cannot explain the difference.
In the study by Sakamoto et al.,  the mean age of his patients was 74.5 ± 7.5 years and no range was given. As his study population was small, it means that the presence of a few extremely low age groups could have brought down the mean age of his perhaps older patient population. If this is true, it could explain why his mean actual testicular volume is lower as testicular volume (though relatively constant after puberty) has been found to start decreasing from the eight decade of life. 
Also, the fact that we studied different population groups; they studied people in Tokyo Japan, and we studied the population in Nnewi Nigeria could account for the difference. As environment and race have also been found to influence testis volume. ,,
The Prader orchidometer overestimated testicular volume by 25.10% (2.66 mls) in this study. It has been shown to overestimate the actual testicular volume by a lot of other studies to varying degrees. ,, This is not unexpected as it adds the volume of the testicular covering. Other factors known to affect Prader orchidometer measurement are not properly separating the epididymis, size of the testis and expertise of the attending physician.  Paltiel et al.  in their study of 18 canine testes showed the orchidometer overestimated testicular volume by only 12% (1.6 ml). The actual cause of this difference is not certain, but it should be noted that he used dog testes and not human testes. Secondly, he did not measure the actual size of testes by water displacement like was done in this study. He got the testicular volume by weighing the testis and converting with the formula, volume = weight/density, while assuming that the density of dog testis is the same for human testis. Though the density of human testis has been found to be constant at 1.038 by Handerlsman et al.,  the assumption by Paltiel et al.  that the density of dog testes should be the same as that of humans is not entirely correct. In addition, the skin of dog scrotum may also have a different texture compared to that of human scrotum.
On the other hand, Sakamoto et al.  found that Prader orchidometer overestimated the mean testicular volume by 81.7% (6.68 ml). In his methodology, though they said the Prader orchidometer measurement was done by one experienced urologist, no mention was made of whether he attempted to exclude the epididymis which is one of the things that has been found to reduce the accuracy of Prader orchidometer. ,,
Rivkees et al.  showed that Prader orchidometer only over estimated by 30%. In his study however, he used animal models. They used the testes of 10 calves and nine dogs, and an artificial scrotum to simulate human scrotum. They also found that Prader orchidometer overestimates testicular volume to varying degrees depending on the size, but no mention was made of the average volume of the testes they measured. The smaller the testis the more the overestimation and vice versa, thus mention of the mean size of the testes measured would have been most appropriate. ,,,
In this study, it was found that Prader orchidometer volume measurements correlated strongly with the actual testicular volumes measured by water displacement using Pearson correlation coefficient (r = 0.926, P = 0.0001). There is no consensus by previous studies on whether or not Prader orchidometer measurement correlates with actual testicular volume measurement. ,, Sakamoto et al.  found a strong correlation between the two (r = 0.818, P = 0.0001). Paltiel et al.  on the other hand did not find a statistically significant correlation between Prader orchidometer and actual testicular volumes (r2 = 0.14, P = 0.12). Though they worked on animals and did not use the actual volume, but a derived one like was highlighted above. Even when Sakamoto et al. agree it correlate as in our own study, the accuracy recorded vary widely when compared to our result (81.7% overestimation as against 25.1% in our study). This may well be because of the inter-observer variability in the use of Prader orchidometer like was found by Tatsunami et al.  Therefore, there will be need for training and standardization for Prader orchidometer use so that its accuracy will be reproducible between attending physicians for clinical and research use.
| References|| |
|1.||Turek PJ. Male infertility. In: Tanagho EA, McAninch JW, editors. Smith's General Urology. 17 th ed. USA: McGraw-Hill Companies Inc; 2008. p. 687-717. |
|2.||Sigman M, Jarrow JP. Male infertility. In: Walsh PC, Retic AB, Vaughan ED, Wein AJ, editors. Campbell's Urology. 8 th ed. Philadelphia: Saunders; 2002. p. 1474-531. |
|3.||Lipshultz LI, Corriere JN Jr. Progressive testicular atrophy in the varicocele patient. J Urol 1977;117:175-6. |
|4.||Sakamoto H, Ogawa Y, Yoshida H. Relationship between testicular volume and testicular function: Comparison of the Prader orchidometric and ultrasonographic measurements in patients with infertility. Asian J Androl 2008;10:319-24. |
|5.||Lenz S, Thomsen JK, Giwercman A, Hertel NT, Hertz J, Skakkebaek NE. Ultrasonic texture and volume of testicles in infertile men. Hum Reprod 1994;9:878-81. |
|6.||Sakamoto H, Yajima T, Nagata M, Okumura T, Suzuki K, Ogawa Y. Relationship between testicular size by ultrasonography and testicular function: Measurement of testicular length, width, and depth in patients with infertility. Int J Urol 2008;15:529-33. |
|7.||Jarow JP, Sharlip ID, Belker AM, Lipshultz LI, Sigman M, Thomas AJ, et al. Best practice policies for male infertility. J Urol 2002;167:2138-44. |
|8.||Costabile RA, Skoog S, Radowich M. Testicular volume assessment in the adolescent with a varicocele. J Urol 1992;147:1348-50. |
|9.||Sayfan J, Siplovich L, Koltun L, Benyamin N. Varicocele treatment in pubertal boys prevents testicular growth arrest. J Urol 1997;157:1456-7. |
|10.||Salzhauer EW, Sokol A, Glassberg KI. Paternity after adolescent varicocele repair. Pediatrics 2004;114:1631-3. |
|11.||David OO, Iyekoretin E. Undescended testes in a developing country: A study of the management of 71 patients. Afr J Paediatr Surg 2008;5:11-4. |
|12.||Zucchi A, Mearini L, Mearini E, Fioretti F, Bini V, Porena M. Varicocele and fertility: Relationship between testicular volume and seminal parameters before and after treatment. J Androl 2006;27:548-51. |
|13.||Marshall WA, Tanner JM. Variations in the pattern of pubertal changes in boys. Arch Dis Child 1970;45:13-23. |
|14.||Kamischke A, Baumgardt A, Horst J, Nieschlag E. Clinical and diagnostic features of patients with suspected Klinefelter syndrome. J Androl 2003;24:41-8. |
|15.||Boisen E. Testicular size and shape of 47, XYY and 47, XXY men in a double-blind, double-matched population survey. Am J Hum Genet 1979;31:697-703. |
|16.||Pitteloud N, Hayes FJ, Dwyer A, Boepple PA, Lee H, Crowley WF Jr. Predictors of outcome of long-term GnRH therapy in men with idiopathic hypogonadotropic hypogonadism. J Clin Endocrinol Metab 2002;87:4128-36. |
|17.||Raivio T, Falardeau J, Dwyer A, Quinton R, Hayes FJ, Hughes VA, et al. Reversal of idiopathic hypogonadotropic hypogonadism. N Engl J Med 2007;357:863-73. |
|18.||Rivkees SA, Hall DA, Boepple PA, Crawford JD. Accuracy and reproducibility of clinical measures of testicular volume. J Pediatr 1987;110:914-7. |
|19.||Paltiel HJ, Diamond DA, Di Canzio J, Zurakowski D, Borer JG, Atala A. Testicular volume: Comparison of orchidometer and US measurements in dogs. Radiology 2002;222:114-9. |
|20.||Sakamoto H, Saito K, Oohta M, Inoue K, Ogawa Y, Yoshida H. Testicular volume measurement: Comparison of ultrasonography, orchidometry, and water displacement. Urology 2007;69:152-7. |
|21.||Diamond DA, Paltiel HJ, DiCanzio J, Zurakowski D, Bauer SB, Atala A, et al. Comparative assessment of pediatric testicular volume: Orchidometer versus ultrasound. J Urol 2000;164:1111-4. |
|22.||Shiraishi K, Takihara H, Kamiryo Y, Naito K. Usefulness and limitation of punched-out orchidometer in testicular volume measurement. Asian J Androl 2005;7:77-80. |
|23.||Ogunbiyi JO, Shittu OB. Increased incidence of prostate cancer in Nigerians. J Natl Med Assoc 1999;91:159-64. |
|24.||Hsieh ML, Huang ST, Huang HC, Chen Y, Hsu YC. The reliability of ultrasonographic measurements for testicular volume assessment: Comparison of three common formulas with true testicular volume. Asian J Androl 2009;11:261-5. |
|25.||Handelsman DJ, Staraj S. Testicular size: The effects of aging, malnutrition, and illness. J Androl 1985;6:144-51. |
|26.||Ku JH, Kim ME, Jeon YS, Lee NK, Park YH. Factors influencing testicular volume in young men: Results of a community-based survey. BJU Int 2002;90:446-50. |
|27.||Main KM, Toppari J, Suomi AM, Kaleva M, Chellakooty M, Schmidt IM, et al. Larger testes and higher inhibin B levels in Finnish than in Danish newborn boys. J Clin Endocrinol Metab 2006;91:2732-7. |
|28.||Rundle AT, Sylvester PE. Measurement of testicular volume. Its application to assessment of maturation, and its use in diagnosis of hypogonadism. Arch Dis Child 1962;37:514-7. |
|29.||Sakamoto H, Saito K, Ogawa Y, Yoshida H. Testicular volume measurements using Prader orchidometer versus ultrasonography in patients with infertility. Urology 2007;69:158-62. |
|30.||Tatsunami S, Matsumiya K, Tsujimura A, Itoh N, Sasao T, Koh E, et al. Inter/intra investigator variation in orchidometric measurements of testicular volume by ten investigators from five institutions. Asian J Androl 2006;8:373-8. |
[Table 1], [Table 2]
|This article has been cited by|
||Assessment of hormonal status in male infertility. An update
| ||Marcio Concepción Zavaleta, José Luis Paz Ibarra, Anthony Ramos-Yataco, Julia Coronado Arroyo, Luis Concepción Urteaga, Pela J. Roseboom, Carlos A. Williams |
| ||Diabetes & Metabolic Syndrome: Clinical Research & Reviews. 2022; : 102447 |
|[Pubmed] | [DOI]|
||Ultrasonographic measurements of the testicular volume in Turkish boys aged 0–8 years and comparison with international references
| ||Edis Çolak, Behzat Özkan |
| ||Journal of Pediatric Endocrinology and Metabolism. 2022; 0(0) |
|[Pubmed] | [DOI]|
||Training using a simulation-based workshop reduces inaccuracies in estimations of testicular volume
| ||Jessica N. Craig,Megan R. Sharman,Ciara G. Fitzgerald,Dominic Wigg,Beth S. Williams,Ellen E. Wilkinson,Neil P. Wright,Joe Langley,Charlotte J. Elder |
| ||Journal of Pediatric Endocrinology and Metabolism. 2020; 0(0) |
|[Pubmed] | [DOI]|
||Testicular volumetry and prediction of daily sperm output in stallions by orchidometry and two- and three-dimensional sonography
| ||S. Pricking,H. Bollwein,K. Spilker,G. Martinsson,A. Schweizer,S. Thomas,H. Oldenhof,H. Sieme |
| ||Theriogenology. 2017; |
|[Pubmed] | [DOI]|
||Reliability of hypertrophy of the contralateral testis in prediction of the status of impalpable testis
| ||Rafik Shalaby,Ahmad Alshamy,Mohamad Abd-Alrazek,Samir Gouda,Mohamad Mahfouz,Mohamad Shahin,Sabri Moussa,Sayed Elhady,Hassan Algallad |
| ||Annals of Pediatric Surgery. 2017; 13(3): 150 |
|[Pubmed] | [DOI]|
||Diagnostic utility of testosterone priming prior to dynamic tests to differentiate constitutional delay in puberty from isolated hypogonadotropic hypogonadism
| ||Suja P. Sukumar,Anil Bhansali,Naresh Sachdeva,Chirag Kamal Ahuja,Ujjwal Gorsi,Kush Dev Singh Jarial,Rama Walia |
| ||Clinical Endocrinology. 2017; 86(5): 717 |
|[Pubmed] | [DOI]|
||Natural history of testicular size in boys with varicoceles
| ||S. Khasnavis,B.A. Kogan |
| ||Journal of Pediatric Urology. 2015; 11(3): 148.e1 |
|[Pubmed] | [DOI]|